Volume 11 Supplement 3
Adipose segmentation in small animals at 7T: a preliminary study
© Moats et al; licensee BioMed Central Ltd. 2010
Published: 01 December 2010
Small animal MRI at 7 Tesla (T) provides a useful tool for adiposity research. For adiposity researchers, separation of fat from surrounding tissues and its subsequent quantitative or semi- quantitative analysis is a key task. This is a relatively new field and a priori it cannot be known which specific biological questions related to fat deposition will be relevant in a specific study. Thus it is impossible to predict what accuracy and what spatial resolution will be required in all cases and even difficult what accuracy and resolution will be useful in most cases. However the pragmatic time constraints and the practical resolution ranges are known for small animal imaging at 7T. Thus we have used known practical constraints to develop a method for fat volume analysis based on an optimized image acquisition and image post processing pair.
We designed a fat segmentation method based on optimizing a variety of factors relevant to small animal imaging at 7T. In contrast to most previously described MRI methods based on signal intensity of T1 weighted image alone, we chose to use parametric images based on Multi-spin multi-echo (MSME) Bruker pulse sequence which has proven to be particularly robust in our laboratory over the last several years. The sequence was optimized on a T1 basis to emphasize the signal. T2 relaxation times can be calculated from the multi echo data and we have done so on a pixel by pixel basis for the initial step in the post processing methodology. The post processing consists of parallel paths. On one hand, the weighted image is precisely divided into different regions with optimized smoothing and segmentation methods; and on the other hand, a confidence image is deduced from the parametric image according to the distribution of relaxation time relationship of typical adipose. With the assistance of the confidence image, a useful software feature was implemented to which enhances the data and in the end results in a more reliable and flexible method for adipose evaluation.
In this paper, we describe how we arrived at our recommended procedures and key aspects of the post-processing steps. The feasibility of the proposed method is tested on both simulated and real data in this preliminary research. A research tool was created to help researchers segment out fat even when the anatomical information is of low quality making it difficult to distinguish between fat and non-fat. In addition, tool is designed to allow the operator to make adjustments to many of the key steps for comparison purposes and to quantitatively assess the difference these changes make. Ultimately our flexible software lets the researcher define key aspects of the fat segmentation and quantification.
Combining the full T2 parametric information with the optimized first echo image information, the research tool enhances the reliability of the results while providing more flexible operations than previous methods. The innovation in the method is to pair an optimized and very specific image acquisition technique to a flexible but tuned image post processing method. The separation of the fat is aided by the confidence distribution of regions produced on a scale relevant to and dictated by practical aspects of MRI at 7T.
Increased adiposity is a risk factor for many diseases. Unfortunately obesity has become a significant risk factor not only in the United States, but also for the entire globe due to increase caloric intake in many regions. Magnetic resonance imaging (MRI) allows researchers to study the physiological and pharmacological effect of obesity in small animal models. Fat volume quantification plays an important role in the research of obesity induced diseases; this paper will focus on a reliable fat separation method utilizing a typical NMR sequence as opposed to a more exotic chemical shift based methods not yet implemented at high field strengths in small animal imaging in this case specifically at 7T. In addition routine sequences offer higher resolution than chemical shift based methods with resolution being an important factor in small animals.
To date various MRI segmentation methods for fat quantification have been reported [1–4]. However, most of these methods depend only on signal intensity in the weighted images. These methods are based on the assumption that fat tissues in MRI correspond to the high intensity, which introduces uncertainty as fat signals do not actually always represent the highest portion of the intensity histogram (e.g. a tissue with higher proton density). Intensity of MRI is a complex function of many factors including the magnetic field strength, image acquisition technique and acquisition parameters. Therefore, the same region of fat in two different MRI scan protocols may not have the same intensity. Finally, magnetic field inhomogeneity may subtly distort most weighted images especially at higher field strengths, resulting in different intensities for the same tissues within the same scan.
In contrast to a weighted image, the parametric image (transverse relaxation time of each pixel) is calculated from the multi-echo images. The decay of the intensity for each pixel is dependent on the physical feature (relaxation time) and should be independent of the acquisition technique at a given field strength. The calculated relaxation times represent biological characteristics of specific tissue, and can serve as the basis of fat separation [5, 6]. Parametric images provide more reliable approximations of pixels contain fat tissue.
Parametric images, however, are easily corrupted by noise and artifacts derived from instrumentation as well as subject animals themselves. In addition, partial volume  occurs in many images, blurring the boundary of different tissues, which interfere with the precise segmentation directly from a parametric map. Dixon methods  address the fat suppression problem based on chemical shift but have never been implemented at 7T in a small animal scanner to our knowledge. Thus these techniques are beyond the scope of this paper.
Our approach described in this paper is based on the combination of the T1 and T2 optimized first echo images(called first image later) and the parametric images based on the multiply echo acquisition. On one hand, the first image is precisely divided into different regions with optimized smoothing and segmentation methods; and on the other hand, a confidence image is calculated from the parametric image according to the distribution of relaxation time relationship of typical fat from phantoms and test animals. With the addition of the confidence image, a useful software tool was created to offer a reliable and flexible method for fat segmentation.
The key techniques used here will be described in detail according to the processing steps including (A) the fitting of parametric image, (B) segmentation of first image, (C) evaluation of phantom and (D) fat extraction.
A: Fitting of parametric image
To measure the transverse relaxation time for each pixel of the parametric image, a curve is fitted to the decay of intensity with increasing TEs . The non-linear least squares is most commonly adopted [10, 11]. For the fitting of fat data, we used the weighted least square method with baseline subtraction and a least point constraint.
After testing both mono and bi-exponential models, the transverse relaxation time measurement of fat appears to satisfy a mono-exponential physical model known as: 
with Tei=i*te and S0 is the pseudo-proton density, which is relative to the true proton density, T1 value and receiver coil response.
a2.Weighted least square
To fit a set of experimental data to the parametric model, a cost function and optimization method was selected.
Where Ii is the scanned intensity of intensity images in ith echo.
Where wi, the weight of ith point, should represent the confidence of the signal. As the low intensity signals are more likely to be noise and according have less certainty, here wi was simply set as proportion to the intensity (wi= Ii).
Second, for the optimization, a Marquardt-Levenberg algorithm, which is specially designed for multi-dimensional non-linear least squares fitting was selected.
B: Image segmentation of first image
Considerable effort has been devoted to the intesnity image with the aim to segmenting the MRI reconstructed image into tissues. Unfortunately, inhomogeneity, bias, edge blurring and other ill-posed problems require complicated mathematical models. However, unlike the previous methods, only regions within one tissue is our goal for the segmentation (i.e. inhomogeneous fat tissue may be divided into several regions). These regions served as the casting board for the confidence image and fat i ultimately determined by the confidence scores.
To improve the precision our morphological segmentation, bilateral filtering and mean shift methods were used and described below. Here, the image obtained in the first echo is chosen due to its higher SNR relative to the other echoes.
MRI filtering was implemented as a preliminary step to decrease the noise, which replaces the signal of a pixel according to the neighbouring pixels. Filtered image If can be regarded as a convolution between original image I0 and kernel K : If = K ⊗ I0. The filter kernels, typically a matrix of size M*M, represents the number of pixels nearby taken into account. Each element ky (i,j∈[1,M]) at different positions represents the weight at a given point.
Linear and nonlinear are two typical filters used in image processing. In linear convolution filters, the weighting coefficients ki,j only takes into consideration the relative position in kernel K, and remains constant throughout the whole image filtering. Nonlinear filters are relative to the target pixel and the coefficients are calculated as a function of local variations of the signal . In the linear filter class, average and Gaussian filers are often used.
Among the nonlinear filters, the median filter is popular. As well, a selective blurring filter  is used [11, 17], which emphases the pixels with similar intensity to the target pixel. A bilateral filter  is an edge preserving technique proposed recently and has been widely used in image processing. In comparison to the selective blurring filter, not only is the intensity similarity taken into account, but also the spatial similarity after processing with a uniform or Gaussian kernel. A comparison of the effects of different filters on the image data is demonstrated later.
b2.Mean shift segments
Where xi are data points around x and the function g() is related to a kernel expression, which defines different influences of xi according to their distance from x. The parameter h is the bandwidth used to control the kernel size.
The shift is iteratively performed until x converges to a stable mode point. Because the similar data points will converge to the same or nearby mode points, all the data points in the first image can be segmented by merging the mode points based on their distances. More details of the mean shift segmentation can be found in 
Compared to conventional MR segmentation methods, mean shift is better for analyzing fat images. Mean shift is a local method making it insensitive to non-uniform intensities. Second, unlike the K-mean or EM methods , the number of clusters does not need to be predefined. In addition, the bandwidth parameters can be adjusted for the different sizes of kernel functions, which provide a flexible way to define the scale according to research objectives. After this segmentation process is complete, the first image will be divided into regions.
The fitted T2 values of fat pixels in a given region are not exactly the same but rather present a distribution. To evaluate this distribution, a phantom study was done.
c1. SNR effect on T2 distribution
From previous literature , it is known that the signal to noise ratio (SNR) can influence this distribution, whether or not variances are caused by the animal or acquisition technique. We excluded variance of individual animals, only the factors related to the acquisition technique were taken into account, which included the effect of the imaging protocol and the instrument.
From the aspect of the MRI protocol, SNR is decided by many parameters, including ratio of echo time, spatial resolution, thickness of slices, and receiver band width. From the instrumentation aspect, it is more complicated to understand the physical factors, which include coil setup, magnetic field bias, RF in homogeneities, and phase deviation. Therefore, to evaluate the relationship between SNR and T2 distribution is a complicated task. A practical way is to determine SNR effects is to use a homogeneous sample with the similar acquisition parameters in the same instrument .
c2. Phantom simulation
Where σ denotes the root mean square (RMS) noise in the background and avg() is the average signal value in lard area.
The noise is assumed to follow a Gaussian distribution when SNR>2 because the actual non-zero average Rician noise will become a quasi-Gaussian distribution in both real and imaginary components .
D: Fat extraction with confidence image
d1. Confidence image
From the phantom evaluation, the measured T2 value in the fat region varies relative to the SNR. Assuming a Gaussian distribution of T2 values, a SNR based Gaussian kernel is utilized here to draw the confidence image. With this kernel, mean and standard deviation need to be determined.
In the T2 parametric image, the T2 histogram shows the different T2 distributions, where different peak mean the different tissues with different biological characters. The T2 value in a peak point means the highest distribution of one region. It is possible to approach the real value. Here, the mode is found by performing a 1-D mean shift method.
With the Gaussian kernel, a weighted filter was used with the calculated T2 value pixel by pixel. For each pixel P, the cast weight wp can be calculated as:
with T2,p is the T2 value of P, T2,m is the mode value P belongs to, and σSNR means the standard deviation value corresponding to current SNR determined in the phantom study.
Therefore, the confidence images are defined by the convolution of T2 and SNR based Gaussian kernel: Ic=Iw ⊗ w.
d2. Fat extraction
The probability of each region belonging to adipose tissue is determined by the confidence scores. Fat can be extracted by setting a predefined or real-time threshold on these confidence scores.
Data were generated on the animal MRI (70/16 Bruker PharmaScan, Germany) with a 16cm diameter bore, field strength 7.05 Tesla, and maximal gradient strength of 400mTesla/m. To acquire intensity images, a Multi-spin multi-echo (MSME) Bruker pulse sequence was used with imaging parameters TR=752ms, NE = 10, TE=8.3ms, FOV=40*40mm, slice thickness=0.9mm.
In Figure 6, it demonstrated that WLS is more insensitive to the number of fitted points than LS method. The WLS method improves the consistency of the calculated T2 values and robustness of final results.
In the top part of the graphical user interface (GUI), the traditional method (with only intensity information) is implemented and shown. Either this image or a modified image which is a composite of the original intensity image and weighted parametric image can be displayed.
In the bottom portion of the screen, fat is separated with based confidence images. To provide operator with a clear perspective, all steps are displayed including smoothing (filtering), region segmentation, confidence image and cast image. Operator implement parametric changes can be visualized real-time at each set. We supply the different filters and region segmentation options, and in the confidence image, probabilities are displayed in the pseudo color map (red mean high probability and blue means low probability). Also, based on confidence image, each region in the cast image is colorized in relation to their confidence scores. The effect of varying the confidence threshold on the size of fat region can be observed in real-time. Quantification results are displayed in the right bottom portion of the window.
Comparison of overlap(%)
From Table1, we find that the performance of WT mouse is obviously lower than OB mouse. This is mainly due to the fact that the WT mouse has relatively less fat, so a small variance in fat will lead to high statistical deviation. Also, the subcutaneous fat in the WT mouse is less obvious, which increase the mismatch of the total fat. The statistical results also reveal that the combination method corresponds more closely to manual method with increasing resolution. In lower resolutions, such as 64*64, the fat tissue is easily mixed with muscle pixels and the partial volume degrades the fat extraction process.
Combining the intensity and parametric images, we described a method of fat separation with the confidence image and proved it to be feasible with real data. However, limitations still exist and further experiments are needed to define and refine the technique.
The combination method only has the capacity to measure the total fat. Recently researchers [26, 27] declared high relationships between the fat distribution and diseases, for example, high levels of visceral adipose tissue have been linked to diabetes. Thus, the different fat types (visceral and subcutaneous) need to be separated for disease related researches. Because subcutaneous fat is always located near the body contour, positional knowledge can be taken into account for further separation. Currently, the method has not been investigated for its ability to discriminate between brown and white fat, which still remains challenging in small animal studies.
In this preliminary research, to test the feasibility of the method, the ground truth for comparison is only based on the in vivo imaging technique. More precise results can be obtained through histopathologic comparison based on the MR Images and sections, which is especially helpful in skinny mouse where adipose is less obvious. To compare the results from different modalities, the co-registration technique is important.
More attention should be paid to the relationship between T2 distribution and different imaging protocols as well as the animal models. Various imaging sequences will be designed by changing the imaging parameters and more animals will be scanned which can be divided into more detailed groups according to their genders and ages.
In conclusion, we report a novel method using both intensity image and T2 parametric image to recognize the adipose tissues on 7T MRI. Proved by experiments on the real data, the method provides a good way to accomplish the adipose separation and may serve as a methodological basis for animal studies.
We would like to thank DR. Simerly’s neouroscience group to supply the data; Mkhitaryan Seda and Davoodian Niyera for the data analysis; Harutyunyan Ira and Karapetyan Gevorg for MRI scans and helpful discussions. Publication of thissupplement was made possible with support from the International Society of Intelligent Biological Medicine (ISIBM).
This article has been published as part of BMC Genomics Volume 11 Supplement 3, 2010: The full contents of the supplement are available online at http://www.biomedcentral.com/1471-2164/11?issue=S3.
- Sacha JP, Cockman Michael D, Dufresne Thomas E, Trokhan Darren: Quantification of regional fat volume in rat MRI. Proceedings of the SPIE Medical Imaging. 2003, 5031: 289-297.View ArticleGoogle Scholar
- Leinhard OD, Johansson A, Rydell J, Smedby O, Nystrom F, Lundberg P, Borga M: Quantitative Abdominal Fat Estimation Using MRI. Proceedings of 19th International Conference on Pattern Recognition. 2008, 1-4.Google Scholar
- Monziolsab M, Guylaine Colleweta, Mariettea F, Koubab M, Davenela A: Muscle and fat quantification in MRI gradient echo images using a partial volume detection method,Application to the characterization of pig belly tissue. Magnetic resonance imaging. 2005, 23: 745-755.View ArticleGoogle Scholar
- Robinson K, Whelan P, Ghita O, Brennan D: Measurement and localisation of body fat in whole body mri. IEI Biomedical Engineering Research Award, (Dublin, Ireland). 2005Google Scholar
- GUTHAUSEN Andreas, GUTHAUSEN Gisela, KAMLOWSKI Andreas, TODT Harald, BURK Wolfgang, SCHMALBEIN Dieter: Measurement of fat content of food with single-sided NMR. Journal of the American Oil Chemists’ Society. 2004, 81: 727-731.View ArticleGoogle Scholar
- Emil Veliyulin: Rapid measurements of fat content in live salmon. [http://www.sintef.no/upload/Fiskeri_og_havbruk/Faktaark/Fat%20content_Eng.pdf]
- Pokric £M, Thacker NA, Scott MLJ, Jackson A: Multi-dimensional Medical Image Segmentation with Partial Voluming. Proceedings of MIUA Conference, Birmingham. 2001Google Scholar
- Jingfei Ma: Dixon techniques for water and fat imaging. Journal of Magnetic Resonance Imaging. 2008, 28: 543-558.View ArticleGoogle Scholar
- Van der Weerd L, Vergeldt FJ, Adrie de Jager P, Van As H: Evaluation of algorithms for analysis of NMR relaxation decay curves. Magn Reson Imaging. 2000, 18: 1151-1158.View ArticlePubMedGoogle Scholar
- DUNN JF, ZAIMWADGHIRI Y: Quantitative magnetic resonance imaging of the mdx mouse model of Duchenne muscular dystrophy. Muscle & nerve. 1999, 22: 1367-1371.View ArticleGoogle Scholar
- Gensanne D, Josse G, Schmitt AM, Lagarde JM, Vincensini D: In vivo visualization of hyaluronic acid injection by high spatial resolution T2 parametric magnetic resonance images. Skin Research and Technology. 2007, 13: 385-389.View ArticlePubMedGoogle Scholar
- Sijbers J, Dden Dekker AJ, Verhoye M, Raman E, Van Dyck D: Optimal estimation of T2 maps from magnitude MR images. Proceedings of the SPIE Medical Imaging. 1998, 3338: 384-390.View ArticleGoogle Scholar
- Marquardt DW: An algorithm for Least-Squares estimation of nonlinear parameters. Journal of the Society for Industrial and Applied Mathematics. 1963, 11: 431-441.View ArticleGoogle Scholar
- Shen J, Shen W, Castan S, Zhang TX: Sum-box technique for fast linear filtering. Signal Process. 2002, 82: 1109-1126.View ArticleGoogle Scholar
- Jean-Marie Bonnya, Odile Boespflug-Tangulyb, Michel Zancac, Jean-Pierre Renou: Multi-exponential analysis of magnitude MR images using a quantitative multispectral edge preserving filter. J. Magn. Reson. 2003, 161: 25-34.View ArticleGoogle Scholar
- Wang DC, Vagnucci AH, Li CC: Gradient inverse weighted smoothing scheme and the evaluation of its performance. Comput. Graph. Imaging Process. 1981, 15: 167-181.View ArticleGoogle Scholar
- Gensanne D, Josse G, Lagarde JM, Vincensini D: A post-processing method for multiexponential spin-spin relaxation analysis of MRI signals. Phys. Med. Biol. 2005. 2005, 50: 3755-3772.View ArticleGoogle Scholar
- Tomasi C, Manduchi R: Bilateral Filtering for Gray and Color Images. Proceedings of the IEEE International Conference on Computer Vision. 1998, 839-Google Scholar
- Fukunaga K, Hostetler L: The estimation of the gradient of a density function, with applications in pattern recognition. IEEE Transactions on Information Theory. 1975, 21: 32-40.View ArticleGoogle Scholar
- Comaniciu D, Meer P: Mean shift: A robust approach towards feature space analysis. IEEE Transactions on Pattern Analysis and Machine Intelligence. 2002, 24: 603-619.View ArticleGoogle Scholar
- Lucia Ballerini: A Segmentation Technique to Determine Fat Content in NMR Images of Beef Meat. IEEE Transactions on Nuclear Science. 2002, 49: 195-199.View ArticleGoogle Scholar
- Barry Prior: Pixel T2 distribution in functional magnetic resonance images of muscle. J Appl Physiol. 1999, 87: 2107-2114.Google Scholar
- Henkelman RM: Measurement of signal intensities in the presence of noise in MR images. Med. Phys. 1985, 12: 232-233.View ArticlePubMedGoogle Scholar
- Yousem DM, Ihmeida I, Quencer R, Atlas SW: Paradoxically decreased signal intensity on post contrast short-TR MR images. AJNR. 1991, 12: 875-880.PubMedGoogle Scholar
- Bonny JM, Zanca M, Boire JY, Veyre A: T2 maximum likelihood estimation from multiple spin-echo magnitude images. Magn Reson Med. 1996, 36: 287-293.View ArticlePubMedGoogle Scholar
- Iacobellis G: Imaging of visceral adipose tissue: an emerging diagnostic tool and therapeutic target. Curr Drug Targets Cardiovasc Haematol Disord. 2005, 5: 345-353.View ArticlePubMedGoogle Scholar
- Ritchie SA, Connell JM: The link between abdominal obesity: metabolic syndrome and cardiovascular disease. Nutr Metab Cardiovasc Dis. 2007, 17: 319-326.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.