Allelotyping of pooled DNA with 250 K SNP microarrays
© Wilkening et al; licensee BioMed Central Ltd. 2007
Received: 13 February 2007
Accepted: 16 March 2007
Published: 16 March 2007
Genotyping technologies for whole genome association studies are now available. To perform such studies to an affordable price, pooled DNA can be used. Recent studies have shown that GeneChip Human Mapping 10 K and 50 K arrays are suitable for the estimation of the allele frequency in pooled DNA. In the present study, we tested the accuracy of the 250 K Nsp array, which is part of the 500 K array set representing 500,568 SNPs. Furthermore, we compared different algorithms to estimate allele frequencies of pooled DNA.
We could confirm that the polynomial based probe specific correction (PPC) was the most accurate method for allele frequency estimation. However, a simple k-correction, using the relative allele signal (RAS) of heterozygous individuals, performed only slightly worse and provided results for more SNPs. Using four replicates of the 250 K array and the k-correction using heterozygous RAS values, we obtained results for 104.141 SNPs. The correlation between estimated and real allele frequency was 0.983 and the average error was 0.046, which was comparable to the results obtained with the 10 K array. Furthermore, we could show how the estimation accuracy depended on the SNP type (average error for A/T SNPs: 0.043 and for G/C SNPs: 0.052).
The combination of DNA pooling and analysis of single nucleotide polymorphisms (SNPs) on high density microarrays is a promising tool for whole genome association studies.
To find new susceptibility loci for complex diseases on the human genome, a high number of case and control samples is required. An old approach with new perspective is the pooling of cases and controls. The larger the number of analyzed SNPs, the more striking are the advantages of a pooling study. With advanced microarray technology it is now possible to analyze SNPs throughout the whole genome. With the Human Mapping 500 K array set from Affymetrix and the BeadChips from Illumina, over 500,000 SNPs can be genotyped on two arrays. Different groups have tested the reliability of Affymetrix microarrays for pooling studies with either the 10 K array [1–6] or the 50 K array [7, 8]. On these arrays, each SNP is interrogated by 40 probes (20 for the plus and 20 on the minus strand). On the 250 K arrays over 90% of the SNPs are represented by only 24 probes (some SNPs are only on the plus or the minus strand). This reduction of probes, as well as the reduction of the feature size from 18 μm (10 K), and 8 μm (50 K) to 5 μm (250 K) could have a negative influence on the outcome of pooling results. To examine if this is true, we tested the Nsp I 250 K array which represents 262.264 SNPs and is part of the 500 K array set. According to the Data Sheet from Affymetrix, over 85% of the human genome is covered by SNPs within 10 kb distance with this array set. If allelotyping of pooled DNA is feasible with these arrays, whole genome association studies including thousands of samples could be performed within a few weeks in a cost-effective manner.
10 K array
Comparison of accuracies of three algorithms
250 K array
Estimation accuracy in the Nsp 250 K array
no. of pool replicates*
no. of "AB" references
minor allele frequency
0.0 – 0.1
0.1 – 0.2
0.2 – 0.3
0.3 – 0.4
0.4 – 0.5
With our data from the 10 K array, we could confirm that from the three tested methods, the PPC algorithm  gave the best estimates. Compared to other methods, this algorithm (a) utilizes the signal intensities from individual probes (not RAS values); (b) it takes only data from the perfect matches; (c) it applies a second degree polynomial for correction of unequal hybridization; and (d) it uses reference information from all three genotypes (AA, AB, BB). Our results suggest that neither of these parameters alone is responsible for the good performance of the PPC algorithm but the combination of all. However, the need for all three genotypes in the reference samples limits the number of SNPs that can be estimated. Another disadvantage of this method is the time consuming computation in Perl and R. This made it impossible to use the algorithm for our 250 K data yet. For the Nsp 250 K array, we used the k-correction with heterozygous RAS values. This algorithm performed only slightly worse than the PPC algorithm. It was the simplest of the tested algorithms and it scored for more SNPs, because homozygous calls were not required. The algorithm proposed by Craig et al. , also uses RAS values and includes reference information of all three genotypes, which should improve the estimation. However, this method gave the worst estimates for our data set. The algorithm used by Kirov et al. with a reported average error of only 0.014 with 10 K arrays might improve the allelotyping accuracy for 250 K arrays. Instead of using heterozygous references, the correction coefficient k is derived from RAS values of a pool with known allele frequencies. This algorithm was not applied here, because it requires a second independent DNA pool with known allele frequencies. Future studies can use our k values (supplied as Additional Material) for allele frequency estimation on the 250 K Nsp arrays. However, results for SNPs with a very low/high frequency in the reference pool may not be reliable. Another approach could be the combination of the PPC algorithm and the algorithm from Kirov et al. where k is calculated from pooled data of all perfectly matching probes. To avoid the use of reference data in a case-control study with pooled samples, it is also possible to directly compare the signal intensities of the perfectly matching probes between cases and controls as shown by Macgregor et al. . In this study, the use of a correction for unequal hybridization signals had only little effect upon the results. However, also slight improvements can be important for the finding of low susceptibility genes in pooling studies.
Despite the reduction of the feature number and feature size, the absolute error between real and estimated allele frequency with the 250 K array was as low as the one for the 10 K array when using Simpson's k-correction. The correlation between real and estimated allele frequency was even higher with the 250 K array, and the standard deviation was lower. However, our results from the 10 K and the 250 K array are not directly comparable, because (a) pools were constructed from different DNA samples, (b) the experimental protocol was different, (c) different scanners were used for both chips, and (d) the software used for data extraction was different.
As shown in Table 2, the accuracy of the allele frequency estimation improved with the number of pool replicates. The absolute error between three and four replicates only decreased by 0.001. Therefore, we assume that the addition of further technical replicates would not essentially improve the accuracy. In our study, we used pools of identical samples. However, for a case-control study, it might be of advantage to use pools of independent samples to capture the variance between the individuals. In this case, an increase of replicates can improve the accuracy. With increasing number of "AB" references, the error decreased to 0.024 when 35 references were present. In our study, the mean error was smaller when the minor allele frequency was higher. This was also true for the 10 K results using the PPC algorithm, which is in contrast to the results published by Brohede et al. , where the best estimates were obtained at minor allele frequencies <0.1. Interestingly, the accuracy of A/T SNPs was found to be significantly worse than the accuracy of G/C SNPs on the 250 K array. This is probably due to the higher affinity of the G-C hydrogen bound compared to the A-T bound. For the stability of the entire hybridization complex, an unspecific hybridization with "A" or "T" is relatively less important than with "G" or "C". Here we analyzed only one of the two 250 K arrays from the 500 K set. The only difference between the two arrays is the cleavage side in the first fragmentation step. Therefore, we assume that both arrays, Nsp and Sty, perform equally well.
Pooling of samples has several disadvantages compared to a case-control study analyzing individual genotypes: (a) Associations which do not result in a significant change of the allele frequency can be overlooked; (b) Measurement errors can lead to false results; (c) Stratification of the population by age, sex, disease subtype, etc. has to be done before the analysis; (d) Haplotype analysis is only possible under certain conditions [10, 11]; and (e) Analysis of gene-gene interactions can not be performed. However, with advancing technologies and algorithms, the mean measurement error can probably be reduced to values < 0.03 [1, 4]. The use of linkage information should improve the likelihood of finding "real" associations and detect false positive SNPs. Taking the HapMap information (Build 35) for the 10 K array, we found ~30% of the SNPs to be linked to its downstream SNP (LOD >3); with the 500 K array set it was ~50%. With this high linkage, the allele frequency of one SNP can be partly explained by the allele frequency of a linked SNP. To take advantage of this fact, two recent publications propose to use p-value combinations in a sliding-window concept [9, 12]. With increasing number of analyzed SNPs and better linkage information most haplotypes can be explained by individual SNPs .
We think that DNA pooling might be a useful and affordable tool to detecting new candidate genes for genetic diseases, especially at a whole genome level. However, this has to be proven in future association studies with pooled DNA.
DNA pooling and microarray analysis
The determination of the DNA concentration in the individual DNA samples was done with PicoGreen reagent (Molecular Probes) using a standard curve of λ-DNA. From each sample, 50 ng genomic DNA was taken for the pool construction. For the 10 K array, we pooled 26 DNA samples that were individually genotyped before with the 10 K array. For the 250 K array we pooled 88 samples from the HapMap CEPH Population, whose genotype information is available at the HapMap homepage . From individual or pooled samples 250 ng DNA was analyzed on the GeneChip Human Mapping 10 K Xba 131 array or the 250 K Nsp array (Affymetrix) according the manufacturers protocols. Four replicates of the same DNA pool from the 10 K and the 250 K array were processed and hybridized on four different days, respectively. Imaging of the microarrays was performed using either the GCS3000 scanner (10 K array) or the upgraded GCS3000-G7 scanner (250 K array) from Affymetrix. Genotype calls and probe intensity data were extracted with the GDAS software using default parameters (10 K array) or the GTYPE software from Affymetrix setting the call threshold for homozygous and heterozygous calls to 0.26 (250 K arrays). For individual DNA, only arrays with a call rate >93% (as guarantied by Affymetrix) were included in the study. For pooled DNA, only arrays with a detection rate (MDR) >97.8% (corresponding to call rate of >93%, see Results) were used for the allele frequency estimation. One array had to be repeated because of its low MDR (96.7%).
Estimation of allele frequency with the 10 K array
On the 10 K array, each SNP is represented by 40 probes each 25 bp of length. The 40 probes are composed of 20 probes perfectly matching the SNP and 20 probes with a 1 bp mismatch. For the 10 K arrays, the analysis software from Affymetrix calculates the "Median Relative Allele Signal" for the forward (RAS1) and the reverse strand (RAS2) which are derived from all 40 probe intensities. Here, we compared three different algorithms, which take either the RAS values or the probe intensities from the 20 perfect matching probes as input. The k-correction proposed by Simpson, et al. uses RAS values (average of RAS1 and RAS2) from heterozygous genotypes . The k-correction proposed by Craig et al. uses RAS values from all three genotypes . For this correction we excluded RAS1 and RAS2 values with standard deviation >1 (SD from 4 pools) and set values <0 and >1 to 0 and 1, respectively. As reference data for the k-corrections (Simpson et al. and Craig et al.) we used RAS values from 34 arrays analyzed with individual DNA in our lab or RAS values from over 3000 arrays on the web page  provided by Craig et al. . The polynomial based probe specific correction (PPC) from Brohede, et al. uses information of the individual perfect match probe pairs from all three genotypes . As reference data for correction, we used 34 arrays previously analyzed in our lab or k-correction data from 26 arrays kindly provided by Jesper Brohede as external reference.
Estimation of allele frequency with the 250 K array
For the 250 K arrays, the k-correction proposed by Simpson, et al. was used to estimate the allele frequencies . Heterozygous RAS values were taken from a set of 56 arrays (all with call rates >93%), which were previously analyzed with individual DNA in our lab. The average RAS values as well as the discrimination scores were calculated from the cell intensity data using the "R" script from Meaburn et al.  which is freely available . We excluded RAS values from the four pools which had discrimination scores < 0.04, as described by Meaburn et al . The discrimination score (DSsnp is a measure of unspecific hybridization used in the 10 K MPAM mapping algorithm (see Affymetrix GeneChip DNA Analysis Software users' guide for detailed information). This score ranges from 0 to 1 with higher scores indicating greater discrimination between perfect match probes and mismatch probes. Individual SNP data for k-correction is supplied as Additional Material, with k derived from heterozygous RAS values (see Additional file 1) and k derived from RAS values of pooled DNA (see Additional file 2).
We thank Jesper Brohede and Leo M. Schalkwyk for their friendly support with the computer scripts and Dagmar Beiße and Sandrine Tchatchou for their help with the microarray analysis.
- Brohede J, Dunne R, McKay JD, Hannan GN: PPC: an algorithm for accurate estimation of SNP allele frequencies in small equimolar pools of DNA using data from high density microarrays. Nucleic Acids Res. 2005, 33 (17): e142-10.1093/nar/gni142.PubMed CentralPubMedView ArticleGoogle Scholar
- Butcher LM, Meaburn E, Knight J, Sham PC, Schalkwyk LC, Craig IW, Plomin R: SNPs, microarrays, and pooled DNA: identification of four loci associated with mild mental impairment in a sample of 6,000 children. Hum Mol Genet. 2005Google Scholar
- Butcher LM, Meaburn E, Liu L, Fernandes C, Hill L, Al-Chalabi A, Plomin R, Schalkwyk L, Craig IW: Genotyping pooled DNA on microarrays: a systematic genome screen of thousands of SNPs in large samples to detect QTLs for complex traits. Behav Genet. 2004, 34 (5): 549-555. 10.1023/B:BEGE.0000038493.26202.d3.PubMedView ArticleGoogle Scholar
- Kirov G, Nikolov I, Georgieva L, Moskvina V, Owen MJ, O'Donovan M C: Pooled DNA genotyping on Affymetrix SNP genotyping arrays. BMC Genomics. 2006, 7 (1): 27-10.1186/1471-2164-7-27.PubMed CentralPubMedView ArticleGoogle Scholar
- Meaburn E, Butcher LM, Liu L, Fernandes C, Hansen V, Al-Chalabi A, Plomin R, Craig I, Schalkwyk LC: Genotyping DNA pools on microarrays: tackling the QTL problem of large samples and large numbers of SNPs. BMC Genomics. 2005, 6 (1): 52-10.1186/1471-2164-6-52.PubMed CentralPubMedView ArticleGoogle Scholar
- Simpson CL, Knight J, Butcher LM, Hansen VK, Meaburn E, Schalkwyk LC, Craig IW, Powell JF, Sham PC, Al-Chalabi A: A central resource for accurate allele frequency estimation from pooled DNA genotyped on DNA microarrays. Nucleic Acids Res. 2005, 33 (3): e25-10.1093/nar/gni028.PubMed CentralPubMedView ArticleGoogle Scholar
- Macgregor S, Visscher PM, Montgomery G: Analysis of pooled DNA samples on high density arrays without prior knowledge of differential hybridization rates. Nucleic Acids Res. 2006, 34 (7): e55-10.1093/nar/gkl136.PubMed CentralPubMedView ArticleGoogle Scholar
- Meaburn E, Butcher LM, Schalkwyk LC, Plomin R: Genotyping pooled DNA using 100K SNP microarrays: a step towards genomewide association scans. Nucleic Acids Res. 2006, 34 (4): e27-10.1093/nar/gnj027.PubMedView ArticleGoogle Scholar
- Craig DW, Huentelman MJ, Hu-Lince D, Zismann VL, Kruer MC, Lee AM, Puffenberger EG, Pearson JM, Stephan DA: Identification of disease causing loci using an array-based genotyping approach on pooled DNA. BMC Genomics. 2005, 6: 138-10.1186/1471-2164-6-138.PubMed CentralPubMedView ArticleGoogle Scholar
- Barratt BJ, Payne F, Rance HE, Nutland S, Todd JA, Clayton DG: Identification of the sources of error in allele frequency estimations from pooled DNA indicates an optimal experimental design. Ann Hum Genet. 2002, 66 (Pt 5-6): 393-405. 10.1046/j.1469-1809.2002.00125.x.PubMedView ArticleGoogle Scholar
- Pfeiffer RM, Rutter JL, Gail MH, Struewing J, Gastwirth JL: Efficiency of DNA pooling to estimate joint allele frequencies and measure linkage disequilibrium. Genet Epidemiol. 2002, 22 (1): 94-102. 10.1002/gepi.1046.PubMedView ArticleGoogle Scholar
- Yang HC, Pan CC, Lin CY, Fann CS: PDA: pooled DNA analyzer. BMC Bioinformatics. 2006, 7 (1): 233-10.1186/1471-2105-7-233.PubMed CentralPubMedView ArticleGoogle Scholar
- Carlson CS, Eberle MA, Rieder MJ, Smith JD, Kruglyak L, Nickerson DA: Additional SNPs and linkage-disequilibrium analyses are necessary for whole-genome association studies in humans. Nat Genet. 2003, 33 (4): 518-521. 10.1038/ng1128.PubMedView ArticleGoogle Scholar
- HapMap, homepage: http://www.hapmap.org/.
- TGEN, Supplementary_Material: http://public.tgen.org/tgen.org/supplementarydata/neurogenomics/supplementarydata/k-correction_values/.
- SGDP, RAS_calculation: http://sgdp.iop.kcl.ac.uk/oleo/affy/.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.